Real-ear measurements by probe microphone. Definition and applications
Keywords:real-ear measurements, audio analyzer, probe microphone, hearing-aid fitting
Real-ear measurements has allowed the audiologist to have a criterion for assessing the adaptation of reliable and valid instruments. The use of these measures in estimating the goodness of adaptation allows us, among other advantages, to take into account individual differences by providing parameters related to the performance of the hearing aid in real ear. In this article we will carry out a review of the main measurements that can be recorded with a real ear audio analyzer and microphone probe and their main applications.
Visibility and Altmetrics
Hadford, E. (1980). The use of a miniature microphone in the ear canal for the verification of hearing aid performance. Ear and Hearing 1, 329. https://doi.org/10.1097/00003446-198011000-00006
Lauridsen, O. y Günthersen, C. (1981). A new probe microphone for investigation of the acoustic of the ear. J Acoust Soc Am 69, 1496. https://doi.org/10.1121/1.385784
Lauridsen, O. y Birk Nielsen, H. (1981). A new computerized method for hearing aid fitting based on measurements at the ear drum. Scan Audiol 10, 109. https://doi.org/10.3109/01050398109076169
Pedersen, B., Lauridsen, O. & Birk Nielsen, H. (1982). Clinical measurement of hearing aid insertion gain. Scan Audiol 11, 181. https://doi.org/10.3109/01050398209076215
Pedersen, B. (1982). Probe placement for sound presurre measurements in the aided ear. Scan Audiol 11, 281. https://doi.org/10.3109/01050398209087481
Ringdahl, A, Leijon, A. & Lidén, G. (1982). A new clinical procedure for hearing aid fitting. Acta Otolaryngol (Stockh), Suppl. 386, 43. https://doi.org/10.3109/00016488209108466
Hadford, E., Leijon A., Lidén, G., Ringdahl, A. & Dahlberg, A.K. (1983). A simplified real ear technique for verifying the maximum output of a hearing aid. Ear and Hearing. 1, 329. https://doi.org/10.1097/00003446-198305000-00002
Leijon, A., Harford, E., Lidén, G., Ringdahl, A. & Dahlberg, A. K. (1983). Audiometric earphone discomfort level and hearing aid SSPL 90 as a measured in the human ear canal. Ear and Hearing. 4(4):185-9. https://doi.org/10.1097/00003446-198307000-00002.
Schore, I., Bilger,R. & Hirsh, I. (1960). Hearing aid evaluation: Reliability of repeated measurments. Journal of Speech and Hearing Research, 25,152-170. https://doi.org/10.1044/jshd.2502.152
Schwartz, D., & Walden, B. (1983). Current status of the clinical hearing aid assessment: A reappraisal of an old philosophy. In D. Konkle & W. Rintelmann (Eds.) Principles of speech audiometry (pp. 321-352).
Beattie, R., & Edgerton, R. (1976). Reliability of monosyllabic discimination test in white noise for differentiating among hearing aids. Journal of Speech and Hearing Disorders, 41, 464-476. https://doi.org/10.1044/jshd.4104.464
Dillon, H., James, A., & Ginis, J. (1997). Client oriented scale of improvement (COSI) and its relationship to several other measures of benefit and satisfaction provided by hearing aids. Journal of the American Academy of Audiology, 8(1), 27-43.
Cox, R., & Alexander, G. (1995). The abbreviated profile of hearing aid benefit. Ear and Hearing, 16, 176-185. https://doi.org/10.1097/00003446-199504000-00005
Byrne, D. (1981). Selective amplification: some psychoacoustic considerations. Pp, 260-285 in: F.H. Bess, B.A. Freeman, and J.S. Sinclair, eds. Amplification in Education. A. G. Bell, Washington, DC.
Byrne, D. (1982). Theoretical approaches for hearing aid selection. Pp. 175-179. In G. Studebacker, and D. Fifield 1974. Evaluation of hearing aid fittings for infants. Br. J. Audiol. 8, 47-54. https://doi.org/10.3109/03005367409074353
Blanco, J. L. (1997). El analizador de audífonos. En: Audioprótesis: Teoría y práctica. Editorial Masson, S.A.
Hawkins, D.B. y Haskell G.B. (1982). A comparison of fuctional Gain and 2 cm3 coupler gain. J. Speech Hear. Disrod. 47, 71-76. https://doi.org/10.1044/jshd.4701.71
Jirsa, R., y Norris W.T. (1978). Relationship of acoustic gain to aided threshold improvement in children. J. Speech Hear. Disrod. 43, 348-352. https://doi.org/10.1044/jshd.4303.378
Larson, V.D., G.A. Studebaker, and R.M. Cox. (1977). Sound levels in a 2cc cavity, a Zwislocki coupler and occluded ear canals. J. Am. Aud. Soc. 3, 63-70.
Wetzell, C., and e. R. Harford (1983). Predictibility of real ear hearing and performance from coupler measurements. Ear Hear. 4, 237-242. https://doi.org/10.1097/00003446-198309000-00003
Zenker F y Barajas JJ (1999) Adaptación de audífonos en función del promedio del espectro de la palabra hablada. Estudio de un caso único. En: Logopedia escolar y clínica. Últimos avances en Evaluación e Intervención. Editor: José Domingo Martín Espino, Madrid, Editorial CEPE, pp. 329-336.
Muelller, H. (1992). Terminology and Procedures. In Muller, Hawkins, Northern (eds.): Probe Microphone Measurements: Hearing Aid Selection and Assesment. (pgs. 41-66). San Diego, CA: Singular Publishing Group Inc.
American National Standards Institute. (1997). Methods of Measurement of Real-Ear Performance Characteristics of Hearing Aids. ANSI S3.46-1997. New York: American National Standards Institute Inc.
Feigin, J., Kopun, J., Stelmachowicz, P., Gorga, M. (1989). Probe-tube microphone measure of ear canal sound pressure levels in infants and children. Ear and Hearing, 10(4): 254-258. https://doi.org/10.1097/00003446-198908000-00008
Martin, H., Munro, K., Langer, D. (1997). Real-ear to coupler differences in chilfrenm with grommets. British Journal of Audiology, 3:23-31. https://doi.org/10.3109/03005364000000009
Scollie, S., Seewald, R., Cornelisse, L., Jenstad, L. (1998). Validity and repeatability of levelindependent HL to SPL transforms. Ear and Hearing, 19(5): 407-413. https://doi.org/10.1097/00003446-199810000-00007
Seewald, R. (1997). Amplification: a child-centered approach. Hearing Journal, 50(3): 61.
Valente, M., Potts, L., Valente, M., Vass, W., Goebel, J. (1994). Intersubject Variability of Real-ear Sound Pressure Level: Conventional and Insert earphones. Journal of the American Academy of Audiology, 5(6): 390-398.
Stemachowicz, P., Lewis, D., Seewlad, R., Hawkins, D. (1990). Complex and puretone signals in the evaluation of hearing aid characteristics. Journal of Speech and Hearing Research, 33: 380-385. https://doi.org/10.1044/jshr.3302.380
- 2021-09-15 (2)
- 2001-09-15 (1)
How to Cite
Copyright (c) 2001 Auditio
This work is licensed under a Creative Commons Attribution 3.0 Unported License.
Articles published after 2020
Authors retain copyright and grant the journal right of first publication with the work simultaneously licensed, unless otherwise indicated, under a Creative Commons Attribution License that allows others to share the work with an acknowledgement of the work's authorship and initial publication in this journal. The authors can choose between the following Creative Commons licenses: https://creativecommons.org/about/cclicenses/
Articles published between 2001 and 2020
The texts published in this journal in the section "AUDITIO 2001-2020" are subject - unless otherwise indicated - to a Creative Commons Attribution 3.0 Spain license. You can copy, distribute, communicate them publicly, make derivative works and commercial uses provided that you acknowledge the credits of the works (authorship, name of the journal, publishing institution) in the manner specified by the authors or by the journal. The full license can be consulted at http://creativecommons.org/licenses/by/3.0/es/deed.es.